تعمیر ، سرویس و خدمات پس از فروش محصولات شرکت Jeio Tech (جیوتک) کره جنوبی
Electric furnace for high temperature experiment at 1100 ℃ / 1500 ℃
Model Name : MF2-12G/GF, MF2-22G/GF, MF2-32G/GF
MF2-12H/HF, MF2-22H/HF, MF2-32H/HF
Enhanced user convenience of touch controller
Intuitive use with a large 5-inch TFT touch controller.
Auto 3 temperature recommendation based on operating history without any recording or storage.
Timer(wait on, wait off) and Ramp(temperature increase rate) function
Monitor and control via a mobile app anytime, anywhere, with LC Connected. (Mobile monitoring system) (Option)
Recording the 36 most recent events, such as start/stop of operation and error occurrence.
Easy program settings intuitively without a manual (MF2-G, MF2-H : program type only)
Available to set up to 10 programs and 10 steps.
Available to set up to 99 repetitions of program from 0.
Function of setting the start temperature to begin an experiment at an accurate temperature the user wants.
Possible to skip to the next step or hold the current step during program operation.
Available to set a deviation of temperature or time to keep going a program operation even if the temperature is not reached to the target accurately.
A useful scheduler function available to select and schedule the desired day, time, and program for repetitive experimental settings.
Highly reliable temperature control performance
Possible to set the rate of temperature change (℃/min) (Ramp function) to control it.
Minimizes heat loss through the use of durable ceramic insulation materials.
Excellent interior sealability due to shock absorbers on the doors and a sturdy double-door structure.
Stable temperature control through digital PID control.
High-precision temperature control over a wide range of temperatures through a 3-point calibration process.
Various safety functions ensure user safety
A safety door lock function that opens only below the door lock temperature set by user.
A built-in electronic overheat protection system that prevents the equipment from being damaged due to overheating.
Front surface of the product is safe even when operating at the maximum temperature.
The equipment stops working when the door is opened for safe use.